Institutional Repository of Peking University School of Basic Medical Sciences
|Application and Development of Review Criteria on Sysmex XT-2100 in Chinese at a Large Third-Grade Class-A Hospital|
|Yue, Jiaxin1; Cong, Yulong1; Wang, Chengbin1; Ma, Junlong1; Ma, Xiaojing2; Lan, Yating1; Xiang, Daijun1|
|关键词||complete blood count microscopic review hematology analyser Sysmex XE 2100|
|WOS标题词||Science & Technology|
|类目[WOS]||Medical Laboratory Technology|
|研究领域[WOS]||Medical Laboratory Technology|
|关键词[WOS]||HEMATOLOGY ANALYZER ; XE-2100|
Background: Hematology analysis is an essential component of patient assessment and used in screening, diagnosis, and the planning of care. The objective of the study was to find out the suitable hematology review criteria for large scale general hospitals in China via the analysis of experimental data from Sysmex XE-2100 hematology analyzer.
Methods: A total 1486 blood samples were detected with the Sysmes XE-2100. Based on hematology review criteria suggested by international consensus group and a positive smear finding new optimal review rules were determined.
Results: With the International Article 41 Review Rules, the true positive ratio (TP), the false positive ratio (FP), the true negative ratio (TN), and the false negative ratio (FN) was 14.0% (208/1486), 31.49% (468/1486) 52.42% (779/1486), and 2.09% (31/1486), respectively. With the help of Laboman 4.2 software (the Sysmex Corporation), 19 rules for review of automated CBC and WBC differential were set up. With our review rules, the TP, FP, TN, and FN was 13.86% (206/1486), 25.17% (374/1486), 58.75% (873/1486) and 2.22% (33/1486), respectively. The review rules were validated, the FN was 0.96%, blasts and immature cells were not omitted.
Conclusions: The review criteria can be developed in light of the rules of the International Consensus Group for Hematology Review, but should be improved depending on different laboratory′s requirements.
|作者单位||1.Chinese Peoples Liberat Army Gen Hosp, Dept Clin Lab, Beijing 100853, Peoples R China|
2.Peking Univ, Sch Basic Med Sci, Dept Clin Med 2008, Beijing 100083, Peoples R China
|Yue, Jiaxin,Cong, Yulong,Wang, Chengbin,et al. Application and Development of Review Criteria on Sysmex XT-2100 in Chinese at a Large Third-Grade Class-A Hospital[J]. CLINICAL LABORATORY,2013,59(7-8):789-794.|
|APA||Yue, Jiaxin.,Cong, Yulong.,Wang, Chengbin.,Ma, Junlong.,Ma, Xiaojing.,...&Xiang, Daijun.(2013).Application and Development of Review Criteria on Sysmex XT-2100 in Chinese at a Large Third-Grade Class-A Hospital.CLINICAL LABORATORY,59(7-8),789-794.|
|MLA||Yue, Jiaxin,et al."Application and Development of Review Criteria on Sysmex XT-2100 in Chinese at a Large Third-Grade Class-A Hospital".CLINICAL LABORATORY 59.7-8(2013):789-794.|